The picnometer should be calibrated with distilled water at 15° C.
The galvanometer can be calibrated with this resistance, which is known as a shunt.
Both instruments must, of course, be calibrated as I have explained before.
The instruments can be so calibrated that it is quite easy to make the alteration.
They should be calibrated by the office of weights and measures.